<?xml version="1.0" encoding="ISO-8859-1"?>

<FDSNStationXML xmlns:xsi="http://www.w3.org/2001/XMLSchema-instance" xmlns="http://www.fdsn.org/xml/station/1" schemaVersion="1.0" xsi:schemaLocation="http://www.fdsn.org/xml/station/1 http://www.fdsn.org/xml/station/fdsn-station-1.0.xsd">
 <Source>IRIS-DMC</Source>
 <Sender>IRIS-DMC</Sender>
 <Module>IRIS WEB SERVICE: fdsnws-station | version: 1.0.7</Module>
 <ModuleURI>https://service.iris.edu/fdsnws/station/1/query?station=MEEK&amp;network=AU&amp;level=response</ModuleURI>
 <Created>2013-11-30T21:14:52</Created>
 <Network code="AU" startDate="1994-01-01T00:00:00" endDate="2500-12-12T23:59:59" restrictedStatus="open">
  <Description>Geoscience Australia</Description>
  <TotalNumberStations>154</TotalNumberStations>
  <SelectedNumberStations>2</SelectedNumberStations>
  <Station code="MEEK" startDate="2003-06-25T00:00:00" endDate="2008-05-12T00:00:00" restrictedStatus="open">
   <Latitude>-26.638</Latitude>
   <Longitude>118.615</Longitude>
   <Elevation>530.0</Elevation>
   <Site>
    <Name>Meekatharra, Western Australia</Name>
   </Site>
   <CreationDate>2003-06-25T00:00:00</CreationDate>
   <TotalNumberChannels>3</TotalNumberChannels>
   <SelectedNumberChannels>1</SelectedNumberChannels>
   <Channel locationCode="  " startDate="2003-06-25T00:00:00" restrictedStatus="open" endDate="2008-05-11T23:59:59" code="SHE">
    <Comment>
     <Value>T4B70 TD4</Value>
    </Comment>
    <Latitude>-26.638</Latitude>
    <Longitude>118.615</Longitude>
    <Elevation>530.0</Elevation>
    <Depth>0.0</Depth>
    <Azimuth>90.0</Azimuth>
    <Dip>0.0</Dip>
    <Type>GEOPHYSICAL</Type>
    <SampleRate>20.0</SampleRate>
    <ClockDrift>5.0E-4</ClockDrift>
    <CalibrationUnits>
     <Name>V</Name>
     <Description>emf in volts</Description>
    </CalibrationUnits>
    <Sensor>
     <Type>Guralp CMG40T_1to100/GDAS</Type>
    </Sensor>
    <Response>
     <InstrumentSensitivity>
      <Value>8.09053E8</Value>
      <Frequency>4.0</Frequency>
      <InputUnits>
       <Name>M/S</Name>
       <Description>velocity in meters per second</Description>
      </InputUnits>
      <OutputUnits>
       <Name>COUNTS</Name>
       <Description>digital counts</Description>
      </OutputUnits>
     </InstrumentSensitivity>
     <Stage number="1">
      <PolesZeros>
       <InputUnits>
        <Name>M/S</Name>
        <Description>velocity in meters per second</Description>
       </InputUnits>
       <OutputUnits>
        <Name>V</Name>
        <Description>emf in volts</Description>
       </OutputUnits>
       <PzTransferFunctionType>LAPLACE (RADIANS/SECOND)</PzTransferFunctionType>
       <NormalizationFactor>911772000000</NormalizationFactor>
       <NormalizationFrequency>4</NormalizationFrequency>
       <Zero number="0">
        <Real>0</Real>
        <Imaginary>0</Imaginary>
       </Zero>
       <Zero number="1">
        <Real>0</Real>
        <Imaginary>0</Imaginary>
       </Zero>
       <Pole number="2">
        <Real>-471.227</Real>
        <Imaginary>0</Imaginary>
       </Pole>
       <Pole number="3">
        <Real>-2199.06</Real>
        <Imaginary>0</Imaginary>
       </Pole>
       <Pole number="4">
        <Real>-4.4421</Real>
        <Imaginary>4.4421</Imaginary>
       </Pole>
       <Pole number="5">
        <Real>-4.4421</Real>
        <Imaginary>-4.4421</Imaginary>
       </Pole>
       <Pole number="6">
        <Real>-391.946</Real>
        <Imaginary>850.672</Imaginary>
       </Pole>
       <Pole number="7">
        <Real>-391.946</Real>
        <Imaginary>-850.672</Imaginary>
       </Pole>
      </PolesZeros>
      <StageGain>
       <Value>2025.14</Value>
       <Frequency>4</Frequency>
      </StageGain>
     </Stage>
     <Stage number="2">
      <StageGain>
       <Value>8</Value>
       <Frequency>4</Frequency>
      </StageGain>
     </Stage>
     <Stage number="3">
      <Coefficients>
       <InputUnits>
        <Name>V</Name>
        <Description>emf in volts</Description>
       </InputUnits>
       <OutputUnits>
        <Name>COUNTS</Name>
        <Description>digital counts</Description>
       </OutputUnits>
       <CfTransferFunctionType>DIGITAL</CfTransferFunctionType>
      </Coefficients>
      <Decimation>
       <InputSampleRate>600</InputSampleRate>
       <Factor>1</Factor>
       <Offset>0</Offset>
       <Delay>0.0</Delay>
       <Correction>0.0</Correction>
      </Decimation>
      <StageGain>
       <Value>49938</Value>
       <Frequency>4</Frequency>
      </StageGain>
     </Stage>
     <Stage number="4">
      <Coefficients>
       <InputUnits>
        <Name>COUNTS</Name>
        <Description>digital counts</Description>
       </InputUnits>
       <OutputUnits>
        <Name>COUNTS</Name>
        <Description>digital counts</Description>
       </OutputUnits>
       <CfTransferFunctionType>DIGITAL</CfTransferFunctionType>
       <Numerator>-.000301355</Numerator>
       <Numerator>-.00110339</Numerator>
       <Numerator>-.00134354</Numerator>
       <Numerator>.000852579</Numerator>
       <Numerator>.00584828</Numerator>
       <Numerator>.00997214</Numerator>
       <Numerator>.00634832</Numerator>
       <Numerator>-.00917189</Numerator>
       <Numerator>-.0304036</Numerator>
       <Numerator>-.0389679</Numerator>
       <Numerator>-.0132799</Numerator>
       <Numerator>.055155</Numerator>
       <Numerator>.149874</Numerator>
       <Numerator>.23326</Numerator>
       <Numerator>.266523</Numerator>
       <Numerator>.23326</Numerator>
       <Numerator>.149874</Numerator>
       <Numerator>.055155</Numerator>
       <Numerator>-.0132799</Numerator>
       <Numerator>-.0389679</Numerator>
       <Numerator>-.0304036</Numerator>
       <Numerator>-.00917189</Numerator>
       <Numerator>.00634832</Numerator>
       <Numerator>.00997214</Numerator>
       <Numerator>.00584828</Numerator>
       <Numerator>.000852579</Numerator>
       <Numerator>-.00134354</Numerator>
       <Numerator>-.00110339</Numerator>
       <Numerator>-.000301355</Numerator>
      </Coefficients>
      <Decimation>
       <InputSampleRate>600</InputSampleRate>
       <Factor>3</Factor>
       <Offset>0</Offset>
       <Delay>0.023333</Delay>
       <Correction>0.023333</Correction>
      </Decimation>
      <StageGain>
       <Value>1</Value>
       <Frequency>4</Frequency>
      </StageGain>
     </Stage>
     <Stage number="5">
      <Coefficients>
       <InputUnits>
        <Name>COUNTS</Name>
        <Description>digital counts</Description>
       </InputUnits>
       <OutputUnits>
        <Name>COUNTS</Name>
        <Description>digital counts</Description>
       </OutputUnits>
       <CfTransferFunctionType>DIGITAL</CfTransferFunctionType>
       <Numerator>.000183216</Numerator>
       <Numerator>.000732864</Numerator>
       <Numerator>.0010993</Numerator>
       <Numerator>.000732864</Numerator>
       <Numerator>.000183216</Numerator>
       <Denominator>.517478</Denominator>
       <Denominator>-2.40934</Denominator>
       <Denominator>4.23886</Denominator>
       <Denominator>-3.34407</Denominator>
       <Denominator>1</Denominator>
      </Coefficients>
      <Decimation>
       <InputSampleRate>200</InputSampleRate>
       <Factor>10</Factor>
       <Offset>0</Offset>
       <Delay>0.01</Delay>
       <Correction>0.01</Correction>
      </Decimation>
      <StageGain>
       <Value>1</Value>
       <Frequency>4</Frequency>
      </StageGain>
     </Stage>
    </Response>
   </Channel>
  </Station>
 </Network>
</FDSNStationXML>
